In flow measurement and fluid dynamics, two popular devices are commonly used to measure flow rates: the Venturi meter and the orifice meter. Though similar in purpose, these instruments differ in their design and operational principles.

## Venturi Meter and Orifice Meter Overview

Flow measurement is a crucial aspect of fluid dynamics, and two popular devices used for this purpose are venturi meters and orifice meters. Both meters are used to measure the flow rate of fluids in pipes by utilizing Bernoulli’s equation and fluid dynamics principles to calculate the flow rate.

### Venturi Meter

The Venturi meter is a flow metering device that consists of a converging section, a throat, and a diverging section. The construction of a typical venturi meter is shown below:

### Elevate Your Engineering With Excel

Advance in Excel with engineering-focused training that equips you with the skills to streamline projects and accelerate your career.

The converging section causes fluid velocity to increase, which results in a decrease in pressure according to Bernoulli’s equation:

As the fluid moves through the converging section and reaches the throat, the pressure decreases while the velocity reaches its maximum value. Once the fluid passes the throat and moves through the diverging section, the pressure starts to recover, and the velocity decreases. The flow rate can be calculated based on the difference in pressure between the entrance and the throat.

Venturi meters have several advantages:

- Low pressure loss
- Wide measuring range
- High accuracy
- Less prone to wear and tear

### Orifice Meter

An Orifice meter is another flow-measuring device that consists of a flat plate with a hole (orifice) in the center, placed perpendicular to the direction of fluid flow in the pipe. As the fluid approaches the orifice, its velocity increases, and the pressure decreases, similar to the Venturi meter. This behavior can also be predicted from Bernoulli’s law.

The flow rate can be calculated using the difference in pressure between the entrance (upstream) and the downstream section (just after the orifice). However, due to the sudden change in the pipe diameter at the orifice, the Orifice meter experiences a much higher pressure loss than the Venturi meter.

Some advantages of Orifice meters are:

- Simple design and construction
- Low cost
- Ease of installation

## Principle of Operation

Venturi and Orifice meters have different operating principles that make each suitable for specific flow measurement applications.

### Venturi Meter

The Venturi meter operates on the principle of the Bernoulli equation, which relates the pressure, kinetic energy, and gravitational potential energy of a fluid flowing through a pipe. As the fluid flows through the converging section of the Venturi meter, its velocity increases, and the pressure decreases. The pressure decreases as it flows through the throat (narrowest part) and then increases again when the fluid passes through the diverging section as the velocity decreases.

The pressure difference between the inlet and throat of the Venturi meter is a measure of the flow rate, and can be calculated using the following equation:

The flow rate through the Venturi meter can be calculated from the following equation:

Where:

- Q is the flow rate
- Cd is the venturi meter discharge coefficient
- A1 and A2 are the cross-sectional areas of the pipe and throat
- P1 and P2 are the pressures at the inlet and throat
- ρ is the fluid density

Discharge coefficients for venturi meters vary with the ratio d/D and range from approximately 0.95 to 1.0 as illustrated in the table below:

### Orifice Meter

The Orifice meter also operates based on the Bernoulli equation. When a fluid flows through an orifice plate with a smaller diameter than the pipe, it experiences an increase in velocity and a decrease in pressure at the orifice’s opening. The pressure drop across the orifice is proportional to the flow rate of the fluid. The difference in pressure upstream and downstream of the orifice can be measured using pressure taps and used to calculate the flow rate using the following equation:

Where:

- Q = flow rate
- C
_{d}= discharge coefficient - A = orifice area
- P1-P2 = pressure difference across the orifice plate
- ρ = fluid density

Discharge coefficients for orifice meters vary with orifice geometry and range from approximately 0.6 to 0.7:

For both Venturi and Orifice meters, the flow rate calculation requires knowing the cross-sectional area of the flow path and the pressure difference across the flow restriction. However, Venturi meters have a more gradual pressure recovery, resulting in higher flow rates with lower pressure losses, whereas Orifice meters are simpler, smaller, and cheaper but have a higher permanent pressure loss and are more susceptible to errors due to wear and erosion.

### Orifice Meter

On the other hand, an Orifice meter consists of a thin orifice plate with a hole in the center, placed perpendicular to the fluid flow direction. As the fluid passes through the orifice plate, its velocity increases, and pressure decreases.

While Orifice meters are generally less accurate than Venturi meters and may result in higher energy losses, they are typically cheaper, easier to install, and require less maintenance. This makes them suitable for applications that do not demand high precision or can withstand a higher pressure drop.

## Key Differences Between Venturi and Orifice Meters

The main differences between the Venturi meter and the Orifice meter are related to their accuracy, pressure drop, installation, maintenance, and cost.

### Accuracy

Venturi meters are known for their higher accuracy compared to orifice meters. This is because the smooth converging and diverging sections in a Venturi meter reduce disturbances and energy loss in the flow stream, resulting in less deviation from theoretical values. In contrast, orifice meters introduce a sharp edge, causing turbulence and a greater chance for measurement errors.

### Pressure Drop

The Venturi meters have a lower pressure drop than orifice meters. The streamlined design of Venturi meters allows the fluid to pass through with minimal resistance, ensuring that most pressure remains unchanged. On the other hand, the sudden constriction in orifice meters creates a significant pressure drop, which can negatively impact the efficiency of pumping systems.

### Installation

Installation of a Venturi meter requires a longer straight pipe section than an orifice meter, mainly due to its design. This can make Venturi meters more challenging to install in space-constrained scenarios. With their simpler and more compact design, orifice meters can be easily installed in pipelines with limited space.

### Maintenance

Venturi meters require less maintenance compared to orifice meters. Due to their smooth design, there is minimal pressure on the pipeline walls, which reduces wear and tear. With their sharp edges, orifice meters are more prone to erosion and require periodic inspections and replacements of the orifice plate to maintain accuracy and efficiency.

### Cost

Orifice meters generally have lower initial costs in comparison to Venturi meters. This is because manufacturing an orifice plate is simpler and requires fewer materials than a Venturi meter. However, considering the higher accuracy, lower pressure drop, and reduced maintenance requirements, Venturi meters may offer better long-term value in certain applications.

## Applications and Industries

Venturi and orifice meters are widely used in various industries for measuring the flow rate of fluids in pipes. These devices are common choices for their simplicity, reliability, and cost-effectiveness.

### Venturi Meter

Venturi meters are used in industries where relatively high accuracy, low pressure loss, and low maintenance are essential for measuring flow rates. These applications often involve larger pipes, higher flow rates, and more expensive or non-abrasive fluids. Examples of industries that utilize venturi meters are:

- Water supply and distribution networks
- Petrochemical and chemical industries
- Industrial process control systems
- Gas industries (natural gas, compressed air, etc.)
- Power generation plants

The venturi meter’s design can handle various flow conditions and is suitable for both liquid and gaseous fluids. The equation commonly used for determining the flow rate using a venturi meter is:

### Orifice Meter

Orifice meters are more suitable for applications where a lower initial cost is important, or a higher pressure drop is acceptable. They are generally used in smaller pipes and for a wide range of fluids, including corrosive or particulate-laden substances. Industries that commonly employ orifice meters include:

- Oil and gas pipeline transportation
- Petrochemical and chemical industries
- Wastewater treatment facilities
- Food and beverage processing plants
- Domestic gas metering